- Pipette tip for filling the microinjection capillaries
- Rack package can be sterilized
- Ideal for recovering surplus solution from the capillary
|Purity grade||Eppendorf Quality|
|Packaging type||Disposable rack|
|Volume range||0.5-20 µL (lt. grey)|
Could you please tell me the order number for the Femtotip adapter?Ordering number for the Femtotip adapter is 5176 208.009.
How is it possible to distinguish the different types of Femtotips by the package?Femtotips are delivered in a silver bag, Femtotips II in a golden.
Which tips are the best for injection into the nucleus of MDCK epithelial cells?We recommend Femtotip II (5242 957.000). They have an inner diameter of 0.5 µm and an outer diameter of 0.7 µm, which is very practical for injection into these cells.
Which material is used for the Capillary Safe?It is made of polysterol.
Which Material is used for the custom tips?Borosilikate
Please tell me the diameter of the TransferTips MDS?The TransferTips MDS have a diameter of 20 µm.
Please tell me the material of the Microloader?The Microloader is made of polypropylene.
Are the Microloaders autoclavable?Yes, the box can be autoclaved up to five times.
Are the Microloaders free of RNase? How can you clean them?The Microloaders are free of RNase because of the heat during production. This was confirmed by a test of the PCR-clean procedure. Because of the small tip diameter, a further cleaning process is not recommended.
On which pipette does the Microloader fit?Eppendorf Pipette Research with variable volume 0.5 - 10 µl (order no. 3111 000.122) andthe Eppendorf pipette Reference with variable volume 0.5-10µl (order no.4910 000.018) and 2.0-20µl (4910 000.026).
Please tell me the length of the Microloader tip?Length approx. 100 mm in total, from conus of pulled tip approx. 65 mm.
Please tell me the outer and inner diameter of the Microloader on the tip?The thin pulled tip of the Microloader has an outer diameter smaller than 0.45 mm