epMotion® 5075 LH is the ideal solution for advanced liquid handling procedures. It offers the same outstanding accuracy and precision as the epMotion 5070, making it an ideal tool for demanding small-volume applications, such as real-time PCR setup, magnetic bead purification and all routine pipetting tasks.
- 12 labware positions
- Automatic exchange of all tools
- Pipetting range from 1 µL to 1,000 µL
- Thermal module option
- Plate gripper option
- Compatible with tubes from 0.2 mL to 50 mL
- Compatible with plates up to 384 wells
- Plate stacking
- Optical sensor* checks the loaded deck
- Detection of tip size and tip quantity
|epMotion® 5075 LH systems|
|Dimensions (W x D x
|43 x 24 x 27 in|
|Weight||187 lb/85 kg|
|Power supply||110-130 V, 50/60 Hz|
|Power consumption||1,000 W|
at 1 µL
at 50 µL
at 1,000 µL
Userguide 006 epMotion 5070 5075 Loading the Invitrog
Loading the Invitrogen E-Gels with the epMotion 5070/5075
|English (US)||221.7 kB||download|
Brochure epMotion 5075 LH VAC I love high performa
I love high performance!
|English (US)||204.46 kB||download|
Brochure epMotion 5070 5075 I love pipetting h
I love pipetting - how about you?
|English (US)||18.85 MB||download|
||English (US)||1.8 MB||download|
Can the dispensing tool pipet concentrated HCl?Yes
With which Liquid type can I pipet DMSO?DMSO can be pipet with the Liquid type "Alcohol 98%", see Application note 155.
Is the epMotion 5075 LH upgradable to VAC or MC version?Yes, it is. The LH version is upgradable to VAC or MC version anytime.
Is it possible to upgrade epMotion 5075 Refreher 1 version to Refresher 2 version?Yes, it's possible. Therefore you need the Software and Hardware Upgrade Set 1, Order no. 5075 851.440.
Is it possible to work with ELISA kits on epMotionYes, see Application note 138.
How can I change the preprogrammed Liquid types?In the parameters of a command you can find the point "Options". Under this point is the"Liquid type". Under "Liquid type" you can click on the point "Change parameter".Then you can change the preprogrammed values till you find the optical parameters for the particular liquids.
Is it possible to work with a PCR-Cooler on epMotion?Yes, it's possible, if you work with the Thermoadapter"Frosty" Order no.5075 789.000
Is it possible to upgrade the epMotion 5075 with a additional Thermomodule?Yes, you can integrate up to 3 Thermomodules in epMotion 5075.
Where can I find the current combination of labware for epMotion 5070/75?You can find it under www.epmotion.com.
What is to do, if my labware is not in the labware list?Contact your local epMotion Specialist..
Are the Dispensing tools autoclavable?Yes.
Is it possible to use Tubes of other manufacturer for epMotion?Yes, it's possible, to work with different epMotion racks in different sizes. Also it's possible to work with rack from other manufacturer, if they are in SBS format.
Can the epMotion automatically pick up plates, transport material and change tools?Yes, the epMotion 5075 can transport plates and other things, also it can change the tools automatically.. The epMotion 5070 is a semi-automatically system and not able to transport labware.
What purpose does the optical sensor serve?The optical sensor has three different functions. 1. To scan number of tips and kind of boxes. 2. Labware recognition on the worktable. 3. Amount of liquid in different tubes.
Is it possible to program methods for epMotion Refresher 1 version with the new epMotion Editor 4.00?No, it isn't possible because the methods are not reverse compatible.compatible
Log files can only be signed immediately after an application run has ended. They cannot be signed later.Do not close the "Signature" dialog for the log file until the last required signature has been completed.
When an application is deleted, the system automatically deletes all ID lists that were created during the execution of this application.Export ID lists in advance as files
If the user minimizes and restores the epBlue window when the simulator is active, the simulator operating controls can no longer be accessed and the simulator cannot be closed. Then, the user must close the epBlue client using the Task Manager.Do not minimize the epBlue window when the simulator is active
If the user has deactivated labware that is used in an application which is currently open, the "Run
TMX: method with sample transfer and TMX mixing and sample transfer command "Keep tips" selected under Options. After the sample transfer, the tool remains in the dispensing position, i.e. with the tips in the plate, and the TMX begins the mixing process.Do not select the "Keep tips“ option if this step will be followed by a TMX command.
If the backup medium (e.g., USB storage medium or network drive) is disconnected during a database backup, epBlue terminates with an unhandled exception.Do not remove the backup medium during the backup.
The length of the labware names is limited to 21 characters. If labware from an earlier version of epBlue A1, A2 or CP is imported and the version includes names with more than 21 characters, the names will be truncated during import.The importer outputs a corresponding text notification so the user can modify his applications.
Making of "It's called epMotion"
It's called epMotion
epMotion® epBlue ID™ - The new software module for epMotion